! THIS PAGE IS NOT MAINTAINED ANYMORE !
Observer pointing with EMIR: 28.7.-4.8.2009 (compiled by GQ)
Observer pointing with EMIR: 1.7.-20.7.2009 (compiled by AS)
Observer pointings for MAMBO2 pool, Feb-Apr 2009
pointing corrections between Apr 21 and Apr 28 (DR)
- pointing corrections between Feb 24 and March 30
Observer pointings for January to March 2009
- pointing corrections between March 23 to March 31 (DR)
2009-03-23, MAMBO2 pointing corrections between 17-03 and 23-03 (GQ). poi_MAMBO2_20090317_20090323_2.eps, poi_MAMBO2_20090317_20090323_1.eps
2009-03-17, MAMBO2 pointing corrections between 10-03 and 17-03 (RA).
2009-02-24, B100 pointing corrections between 17-02 and 24-02 (DR).
2009-02-17, pointing corrections between 10-02 and 17-02 without the ones with the encoder problems (JS).
2009-01-24, observer pointings between 14-01 to 24-01 with breaks due to snow storms, no model fit (DR)
2009-01-13, updated by JP, used 35 pointing scans with the B100 receiver. Only the pointing offset parameters P2 and P7 have been modified. The rms previous and after the fit was 1.9". The coverage and scatter of the pointing session is available in PointModel090113.pdf .
2009-01-07, update by JP, 16 points, previous rms 9.5" and new rms = 3.4". point_model090108.pdf . This fit is confirmed by a fit by CK to 89 of in total 231 high-quality observers pointings which were conducted since beginning of this year:
Last pointing models
Pointing model carried out with EMIR, pixel E0V, on 2009-05-26, 35 observed points, previous rms = 2.88", new rms = 1.79". The coverage and scatter of the pointing session were: