NIKA November 02, 2015 / S12
Back to list of day to day logbook of observations
IMPORTANT NOTES ON OBSERVING INFO
Comments: configuration of the acquisition soft all 3 arrays:run13.ini --> the data are saved in /home/nika2/NIKA/Data/run13_all_X
Start the observations at 5a.m
We modified the clean_data_2 to get the median of the TOI on a bigger sample: n_median = round(2*max(!nika.fwhm_nom)/med_speed*!nika.f_sampling) is now changed in n_median = round(4*max(!nika.fwhm_nom)/med_speed*!nika.f_sampling)
--> We get the following results for the beams fwhm: fwhm_1 = 14.3 fwhm_2 = 19.5 fwhm_3 = 14.7 --> the beam is more consistent at 2mm
The flux on Mars is negative on array 3 (bug in the acquisition software) so we changed the sign of the TOI for this array in nk_getdata.